|High temperature rotary furnace
|The opening structure on the double-layer shell makes it easy to take and place the furnace tube. Equipped with an air cooling system
|Furnace tube size approx.
|outer diameter φ80mmX1200mm
|Furnace tube material
|Equal diameter high purity quartz tube
|Heating zone length
|440mm ，The length of the constant temperature zone is approx. 200mm
|The temperature is not greater than 1100℃ under continuous operation; maximum temperature: 1200℃
|Temperature control method
|Adopt PID control and self-tuning adjustment, intelligent 30-segment programmable control
|Ultra high temperature alarm
|With ultra-high temperature alarm
|Broken couple tips
|There is a reminder of couple breakup
|Furnace chamber size
|High purity alumina refractory fiber
|0--10℃/min Adjustable (recommended ≤5℃/min for temperatures above 1000 degrees)
|High quality resistance wire
|Temperature measuring element
|K type thermocouple
|Working power supply
|220V 50HZ 3KW
|Adjustable up to 10 degrees, the angle can be displayed through a digital inclinometer
|Furnace tube speed
|Maximum 10 rpm (speed adjustable, rotation direction adjustable)
|Stainless steel metal flange, with valve and related interfaces, easy to disassemble and assemble, can be evacuated and filled with protective gas to slightly positive pressure
|The lower box of the equipment is a movable box that can also be used as a material rack, making the furnace as a whole easy to move.
|A dynamic and static sealing device and a condensation collection tank are installed on the right side.
Install heating tape between flange and collection tank.
|15 months from the date of shipment or 12 months from the date of acceptance (whichever comes first), consumable parts and man-made damage are not covered by the warranty.
|1. The air pressure inside the furnace tube is less than 0.02MPa
2. When introducing gas into the furnace tube, a pressure reducing valve should be installed on the gas cylinder, which will be more accurate and safer to use.
3. When the equipment is running, it is not recommended to close the air outlet valve and air inlet valve at the flange end of the furnace tube. If you need to close the air valve to heat the sample, you need to always pay attention to the pressure gauge indication. If the air pressure indication is greater than 0.02MPa, you must open the air outlet valve immediately to prevent accidents (such as furnace tube rupture, flange flying out, etc.)
4. When the furnace is at high temperature, the furnace tube cannot be in a vacuum state. The air pressure in the furnace tube must be equal to the atmospheric pressure and maintained at normal pressure.
5. Customers are not recommended to use flammable, combustible, corrosive, or toxic materials or gases. If the process must be used, please install relevant safety devices yourself. The company is not responsible for the problems caused.